View and Download Furuno GP operator’s manual online. DGPS NAVIGATOR GPS NAVIGATOR. GP Marine GPS System pdf manual download. Also for. Furuno GP Pdf User Manuals. View online or download Furuno GP Operator’s Manual. User manual for the device Furuno GP Online user manual database.
|Published (Last):||18 June 2004|
|PDF File Size:||10.9 Mb|
|ePub File Size:||7.33 Mb|
|Price:||Free* [*Free Regsitration Required]|
Page of 88 Go.
Furuno GP-32 Operator’s Manual
GP system configuration GP system configuration Do not disassemble or modify the equipment. Receiver status See table on page 2.
Your unit can automatically advance to the next waypoint on a route, so you do not have to change the destination waypoint repeatedly.
Be sure the power supply If the fuse blows again after replacement, request service. Repeat steps 1 through 5 to complete the route. The fuse holder contains a spring that fixes the fuse. For your convenience If looking through the Furuno GP32 user manual directly on this website is not convenient for you, there are two possible solutions: Below you will find previews of firuno content of the user manuals presented on the following pages to Furuno GP Press the [ENT] key to manuwl your complete menu tree appears in the Appendix.
Press the [ENT] key. Prompt for erasure of track 4. Position Display Format 7. Majual of the content on the page No. Table of Contents Add to my manuals Add. Dgps Setup GPS, in order of availability. Press the [MENU] key once or twice mannual open the menu. Below is a quick AUTO introduction to how to choose a menu and change menu settings. The example screens shown in this 1.
” GPS/WAAS NAVIGATOR GP | GPS, Chart Plotter | Products | FURUNO
Using the online preview, you can quickly view the contents and go to the page where you will find the solution to your problem with Fueuno GP CNT shows the number of times the test has been executed consecutively. Incorrect power supply may cause the equipment to overheat. This equipment is intended for use on marine vessels.
Downloading to your computer – You can also download the user manual Furuno GP32 to your computer and keep it in your files. Odometer distance alarm violated. If you selected 4.
Waypoint attribute edit window 8. Press the [DISP] key to choose a display mode. Waypoints can be entered on the plotter display three ways: Do not use it in other applications.
Voltage of battery in display unit is low. Try to clear backup data. Do not use it in other applications. Many people prefer to read the documents not on the screen, but in the printed version.
The message shown below appears. Speedomete Summary of the content on the page No.
” GPS NAVIGATOR GP | GPS, Chart Plotter | Products | FURUNO
To stop the test, turn off the power. Maanual your reference, a 6. If necessary, change waypoint, position, mark shape, and comment date and time. Receiver status See table on page 2. Enter the alarm range 0.
Furuno GP-32 GPS Receiver User Manual
Be sure the power supply is compatible with the equipment. The horizontal range in the plotter display is available among 0. Previous destination is cancelled whenever a destination is newly set. Power source voltage SOG: Mamual the [MENU] key once To start viewing the user manual Furuno GP32 on full screen, use the button Fullscreen.
Zoom, centering, or escapes from current opera- tion, depending on display in use. The unit beeps The time remaining until the power is turned and then starts up with the last-used display off is counted down furujo the display. Data 2 Output Data: Installation Of Antenna Unit For fixing by post or pipe, it is recommended fuuruno use stays to secure the post or pipe to prevent damage to the GPS receiver by vibration. Initial Settings Autopilot data XTE gp3 bearing to waypoint Bearing from own ship to destination Range and bearing to waypoint great circle navigation GPS position fixing condition time of fix, latitude, longitude, receiving condition, number of satellites used, DOP Latitude and longitude Loran-C time difference Be s Summary of the content on the page No.
Power source voltage too high.