ASTM D – Standard Test Method for Dielectric Breakdown Voltage of Insulating Oils of Petroleum Origin. Using VDE Electrodes. ▫ IEC Insulating. Page 1. Page 2. Page 3. Page 4. Find the most up-to-date version of ASTM D at Engineering
|Published (Last):||5 September 2011|
|PDF File Size:||6.46 Mb|
|ePub File Size:||16.79 Mb|
|Price:||Free* [*Free Regsitration Required]|
Power by seonoco All Rights Reserved: Link to Active This link will always route to the current Active version of the standard. For referee tests or tests that will be used for close comparisons, the laboratories shall agree in advance on the spacing for the tests and ensure that all other requirements of this test method are met.
Automatic equipment should be used to control the voltage rate of rise because of the difficulty of maintaining a uniform voltage rise manually. The sample shall be taken in a dry, clean, non-permeable bottle.
High Voltage Inc TCD-5 Test Cell 60KV – ASTM D1816 test specs
Email Us or call toll-free: Avoid touching the electrodes or the inside of the cell after cleaning. Referenced Documents purchase separately The documents listed below are referenced within the subject standard but are not provided as part of the standard. Available request a quote.
This test method should be used as recommended by professional awtm standards such as IEEE C The equipment should produce a straight-line voltage-time curve over the operating range of the equipment. What are your requirements. It d1186 the responsibility of the user of this standard to establish appropriate safety and health practices and determine the applicability of regulatory limitations prior to use.
Flat “go” and “no-go” gages may be substituted having thicknesses of the specified value more or less 0.
Hipotronics TCVDE Oil Dielectric Test Cell for ASTM D1816 testing
Clearance from the electrodes to all sides, bottom, cover or baffle, and any part of the stirring device is at least Product Description The majority of high-voltage transformers, cables, switchgears, transducers, capacitors, and rectifiers use insulating oils for insulating electrically live parts and to carry off thermal energy.
If it is necessary to readjust the electrodes, lock the electrodes and check the spacing. ASTM Method being used?
Work Item s – proposed revisions of this standard. Pharmaceutical laboratory apparatus stability chambers, tablet disintegration, tablet dissolution, tablet hardness, V-blenders, It is recommended that the circuit design limit the disruptive current duration and magnitude to low values that will minimize damage to the electrodes and limit formation of non-soluble materials resulting 1d816 the breakdown, adtm consistent with the requirements of 4.
Lab Equipment for Sale: After thorough cleaning, flush the cell with new oil of the type to be tested that is filtered through a 5-micron filter and containing less than 25 ppm moisture.
Automatic Oil Tester kV, ASTM D, ASTM D, ASTM D, IP
qstm At the beginning of each days atm, clean according to 7. Up to 75kV rms symmetrical Voltage measurement accuracy: Refer to Terminology D for definitions used in this test method. If a dielectric breakdown does not occur during any of the consecutive breakdown tests using the 2 mm spacing or the sample is not adequate for the 2 mm spacing test cell a 1 more or less 0.
Conduct a voltage breakdown test on a specimen of this oil in the manner specified in this test method.
Calibration s shall be traceable to national standards and calibration should be verified annually or more often to ensure accuracy requirements. Determination of the dielectric breakdown voltage of insulating oils provides an early detection method for any sstm in the insulating properties.
After careful inspection, any electrodes from which pitting cannot be removed by light buffing should be discarded, as more refinishing would destroy the electrode contour and dimensions shown in Fig. The most important requirement of an insulating oil is a astk dielectric strength. This test method is applicable to insulating liquids commonly used in cables, transformers, oil circuit breakers, and similar apparatus as an insulating and cooling medium. Petroleum Products and Biofuel lab apparatus baths, calorimeters, color measurement, astj meters, flash point, hydrogenators, oil testing, penetrometers, surface tension, vapor pressure, viscometers and rheometers, In order to preclude stirring air with the sample, provide the cell with a cover or baffle that will effectively prevent air from contacting the circulating liquid.
Phenix Technologies – Phenix Technologies TC/VDE Accessory For Astm D – John Morris
So that the breakdown may be observed, transparent materials are desirable, but not essential. A test cell having a capacity of 0. Calibrate and label automatic controls in terms of rate-of-rise. Historical Version s – view previous versions of standard.
The test cell shall be designed to permit easy removal of the electrodes for cleaning and polishing, verification that the shape is within the specified tolerance, and to permit easy adjustment of the gap spacing.
It is important to avoid touching the electrodes or the cleaned gage with the fingers or with portions of the tissue paper or chamois that have been in contact with the hands.